You have no items to compare.
- Babysense2 is included as a medical device on the Australian Register of Therapeutic Goods (ARTG No. 97479)
- Continually monitors the breathing movement rate of your baby
- An audible alarm is activated if your baby’s breathing movement slows to a rate of less than 10 breaths per minute or cannot be detected for more than 20 seconds
- The control unit connects to sensor pads placed under your baby’s mattress
- Battery operated
Secure710 Key Features:
- 2.4" colour LCD display
- Parent Talk Back
- 2 x Digital Zoom
- Infrared camera for night vision
- Nursery room temperature sensor and alerts
- Multi-camera capable - connects to up to four cameras.
- Additional cameras can be purchased separately
ALWAYS READ THE LABEL AND USER GUIDE, USE ONLY AS DIRECTED.
- Powered by 4 size AA, 1.5 volt hight quality Alkaline batteries (not supplied). Power consumption approx. 1mA.
- Battery Life approx. 6 months using high quality alkaline batteries.
- Alarm volume typically 85dB SPL at a distance of once meter from control unit.
- Size of sensor pad is 216mm in diameter.
FAQ’s for Oricom Babysense2
Re-registering your SC710 Baby Unit (Camera) or registering a new Baby Unit
Switch on the parent unit.
On monitor mode, press and hold <Camera> key for 5 seconds, it will switch to registration mode.
After registration icon appears, press the arrow up or arrow down button to select the camera channel [C1] or [C2] or [C3 or [C4] that you want the baby unit to pair with.
Press and release <Menu> key to confirm which camera number to be registered.
The Link LED then flashes rapidly for 10 seconds.
Switch the baby unit ON. Press the pair key once (at the back of the camera).
If registration is completed, camera video appears on LCD screen and returns to normal mode. The baby unit will also return to normal mode.
If the registration time is expired or registration fails, [Out of Range] appears on the LCD.
Q. Is it a medical device?
A. Babysense2 is included as a medical device on the Australian Register of Therapeutic goods
(ARTG No. 97479).
Q. Is it simple to set up and easy to use?
A. Yes, extremely simple to set up and operate, however we strongly recommend the user guide instructions be followed and functionality regularly tested.
Q. Is it portable?
A. Yes, it uses 4 AA Alkaline batteries (not supplied) so you can move it easily.
Q. I've read that the babysense2 monitor runs on batteries. How long (on average) will the batteries normally last before needing to change them?
A. The babysense2 is Powered by 4 size AA, 1.5 volt Alkaline batteries (not supplied). Power consumption is approximately 1mA. Battery life is approximately 6 months using high quality alkaline batteries.
Q. Can it be used in cribs, bassinets, travel cot?
A. Yes, as long as the base is rigid and a suitable mattress is used and you must ensure that excess cords are secured under the mattress out of reach of baby or other infants.
Q. How many sensor pads does it have?
A. It comes with two highly sensitive sensor pads that cover the area of a standard sized crib or cot.
Q. What age groups is it suitable for?
A. It is suitable for healthy infants up to one year (12months) of age (it cannot be used for children or adults).
Q. If I have twins can I use with both babies in one cot?
A. No you will need to have babies in separate cots each with its own Babysense monitor.
Q. Lights on Control unit do not illuminate when turned ON?
A. Batteries are not inserted correctly or may be flat. Check batteries are inserted in the Control unit observing the correct polarity. Control unit requires 4 x AA 1.5V Alkaline batteries (not supplied).
Q. Low battery voltage warning (red light) on the Control Unit is flashing?
A. Batteries are low, replace all 4 batteries.
Q. False alarm up to 30 seconds from turning on?
A. Single sensor pad connected through the connecting cable.
Sensor pad is defective.
Connecting cable is defective (when using 2 sensors).
Two sensors are not connected correctly to the Control unit.
Cable is connected to the control units blocked socket.
When using only one sensor pad, make sure it is connected directly into the Control unit without the connecting cable. Connect one sensor pad without the connecting cable and lay the baby on the bed for 1minute. If alarm does not activate, the sensor pad is ok. If alarm activates, disconnect and place the sensor pad aside.
Connect the other sensor pad and repeat steps above.
If both sensor pads are in working order, connect the connecting cable and lay the baby on the bed for 1 minute. If the alarm activates the connecting cable is damaged.
Do not connect any cables to the blocked socket on Control unit (only to “SENSOR IN” socket).
Q. Alarm goes off after a few hours
A. The mattress is not touching the sensor pads.
The mattress is raised (there is an object near the sensor pads).
Two sensor pads are connected but the baby is not yet crawling.
Only one sensor pad is connected and the baby has crawled off the sensor pad.
The mattress does not enable adequate transfer of baby movement (a rigid mattress, a mattress with springs or mattress with air pockets).
Make sure that the mattress and sensor pads are touching.
Make sure that there are no objects next to the sensor pads and that the Twin connector is not placed on or near the sensor pads (it is preferable to locate the Twin connector outside the bed area).
Connect sensor pads as per the user manual. If baby is not yet able to crawl, use only one sensor pad – and make sure sensor is placed underneath baby's location in bed to ensure greater relative sensitivity.
Place a sheet of plywood (not supplied) under the sensor pad so that the sensor pad and the mattress will be tightly pressed together.
Q. Delayed alarms while taking the baby out of the bed
A. Performance test is not correct.
Do not conduct performance test by taking the baby out of bed and waiting for the alarm to go off. Kindly conduct performance test according to the performance test instruction in user guide.
The alarm does not go off within 20 seconds when performing the correct performance test
Environmental Influences detected by the sensor pads, such as wind, fan, air-conditioner, toy mobile vibrations.
Perform a test which will detect the source of the movement that the device detects, eg; sources of wind such as air conditioner, an open window or a fan, movements of baby mobile (which can vibrate even if it’s not turned on) walking around the bed area or an intercom placed near the bed with intense transmission levels. Eliminate sources of wind, remove the mobile etc. After neutralizing all environmental interruptions, perform the test again.
Can I use the Babysense2 on my 13 month old infant?
Babysense2 is intended for healthy newborn infants from birth and up to 1 year of age.
Babysense is set to recognize a defined breathing movement rate of infants within this age group (the alarm sounds if breathing stops for 20 seconds or if the breathing rate slows to below 10 breaths per minute).
Q. Can I use Babysense2 if my infant needs constant medical supervision?
A. No. You should contact your GP or medical specialist and seek their advice on what alternative suitable methods of monitoring your infant are available.
Q. Why is it recommended that only one pad be used in a bassinette?
A.The Babysense2 detects the movement produced by a baby’s breathing. A newborn, being smaller, may not have as much movement than a 3 or 4 month old baby, and when two sensor pads are connected it does reduce the sensitivity of the pads, a little, no problem for an older baby but maybe a little more difficult to detect the movement for a new born baby. This is why it is always best to follow the instructions in the user guide including performing a regular (daily) test of the monitor.
Babysense2 is intended for use as an Infant Apnoea Alarm.
ALWAYS READ THE LABEL AND USER GUIDE, USE ONLY AS DIRECTED.
Great Review by Speedy
Price Value Quality
Great (Posted on 22/11/15)
Excellent life saving device Review by sam
Quality Price Value
Can not rate this highly enough. We had our movement alarm sound at 1:30am when our bub was 7 wks old. He had stopped breathing, we were incredibly lucky and blessed that we were able to save his life. I hate to think about what would have happened had we not had this product. Thankyou forever to this invention!!
(Posted on 10/11/15)
Write Your Own Review
Thank you for taking the time to write your review. Your feed back is important to us, good or bad.
Oricom International Pty Ltd